The DY is calibrated according to methods defined in ISO EN 7500-1. It is not possible to calibrate the DY with static loads applied. Instead it...
The DY is calibrated according to methods defined in ISO EN 7500-1. It is not possible to calibrate the DY with static loads applied. Instead it is calibrated using a slowly increasing load. At the time of launch this method was defined in an appendix to the standard. It is now described in the main body of the standard.
In order to carry out this calibration, the software necessary to do it is provided free of charge in the DY-Link software that is available in the downloads section.
For calibration institutes, there is also a video available showing the complete calibration procedure. Please contact your nearest Screening Eagle / Proceq representative to obtain a copy of the video.
Proceq is now part of Screening Eagle Technologies.
Screening Eagle is a merger of Dreamlab, a Singapore-based software and robotics company and Proceq, a Swiss-based NDT company with a 65+ year heritage as a market leader in portable sensors. Together, we protect the built world with software, sensors and data.
You are being redirected to ScreeningEagle.com